I am running the 8890 GC, with 7010B TQ. I am analyzing PCBs and Pesticides on sediment samples. I am having trouble with my endrin breakdown.
After cleaning my inlet, installing a new column, new liner, etc. my Endrin breakdown is still high, way above 20%.
Does anyone have any recommendations as to conditions, liners, etc. that could help with this problem.