I am working whit GC/MSD and the Headspace autosampler (G1888 Agilent).
I have some carryover on the autosampler. I already run whit a higher temperature on the loop and the line and the issue continues.
Could you recommend some maintenance for the autosampler?
Do I need to replace the needle and the loop?