when we are trying injec VOC, this image appear and the frase in system says: "Syringe volume out of range"
Have you rebuilt the method from the default method? This would rule out corruption as the cause of the issue. The error code refers to a plunger position error. This could be the result of a dirty syringe or a hardware issue with the injection unit or you are trying to inject too fast. Given that this method has run for awhile I would check the syringe or you could possibly need service on your injection unit.