. 1. recently, we changed the tubing for internal standard and sometimes found, RSD% for internal standard counts was beyond 5% which raises red flags ( see the below picture)
2. These warnings showed once after 6 or 9 injections. we noticed some very small bubbles were created and accumulated when the sampler was rinsing.
3. what should we do to make internal standard reading consistent?