Have you tried running the built in tests in Lab Advisor for the detectors? Self test? With and without the flow cell.
What values are you getting with the cell test. Please make sure you have 100% water in the flow cell.
You might set the "margin for negative absorbance" higher.. Otherwise ensure you do a balance before doing detection, so the detector does not deliver too negative values to the datasystem.