Log in to follow, share, and participate in this community. Not a member? Join Now! Good afternoon I am working with tuberous root to know monosaccharides, the sample does not have any types of processing ( enzymatic hydrolysis). An Agilent 1260 Infinity with IR detector and Agilent Hi-Plex H... Hi community! I need know HSA Peptide Standard Mix Kit composition (G2455-85001). On the Agilent web site in Safety Data SheetsI don't found information about composition. Thanks! Hi community! Help me please, this G4204-81090 Low pressure mixer assembly for Agilent 1290 Infinity Quaternary Pump is a replacement G4204-60090? I enclosing photo of the requi... En el adjunto explico todo el problema con fotos. We have agilent 1100 series (HPLC with vacuum degasser, binary pump, autosampler, thermostated column compartment and MWD). UV & Vis were on, along with the pump, 30 mins before the run. Baseline was p... Looks like the needle is stuck in the 'up position' as it does not move and the error only occurs when I try to start a run or tell it to reset the needle. I need to make sure that if the power cuts out the systems are able to shut down properly and save whatever experiments were being run in order to prevent data loss.